Wika Pressure Transmitter S 10
Wika Pressure Transmitter S 10:
Wika Pressure Transmitter S 10 for general industrial applications is the ideal solution for customers with demanding measuring requirements. It features a very good accuracy, a robust design and an exceptional number of variants, meaning it can be suited to the widest range of applications.
- Measuring ranges from 0 … 0.1 to 0 … 1,000 bar
- Non-linearity 0.2 % of span (BFSL)
- Output signals: 4 … 20 mA, DC 0 … 10 V, DC 0 … 5 V and others
- Electrical connections: Angular connector form A, circular connector M12 x 1, various cable outlets and others
- Zero point and span adjustable via internal potentiometer
Remove the protection cap by hand only just before installation and absolutely avoid any damage to the diaphragm during installation as well (S-11).
For Model S-10 you have to provide for a sealing element; exceptions are instruments with self-sealing threads (e.g. NPT thread). For Model S-11 the sealing is included in
Please refer to data sheet “Pressure gauge sealing washers AC 09.08” in WIKA’s product catalog Pressure and Temperature Measurement or our website www.wika.de for details about sealing washers.
When mounting the instrument, ensure that the sealing faces of the instrument and the measuring point are clean and undamaged.
Screw in or unscrew the instrument only via the flats using a suitable tool and the prescribed torque. The appropriate torque depends on the dimension of the pressure
connection and on the sealing element used (form/material).
Do not use the case as working surface for screwing in or unscrewing the instrument.
When screwing the transmitter in, ensure that the threads are not jammed.
Connect the instrument to earth via the pressure connection.
For transmitter with voltage output signal: Operate the pressure transmitter with a shielded cable and earth the shield at least on one side of the cable, if the cable is longer than 30 m or if it is run outside of the building.
For North America: The connection shall be made to „Class 2 Circuits“ or „Class 2 Power Units“ according to CEC (Canadian Electrical Code) or NEC (National Electrical Code).
Ingress protection per IEC 60529 (The ingress protection classes specified only apply while the pressure transmitter is connected with female connectors that provide the corresponding ingress protection).
Ensure that the cable diameter you select fits to the cable gland of the connector.
Ensure that the cable gland of the mounted connector is positioned correctly and that the sealings are available and undamaged. Tighten the threaded connection and check the correct position of the sealings in order to ensure the ingress protection.
Please make sure that the ends of cables with flying leads do not allow any ingress of moisture.